GTS meeting 12 20 16
Revision as of 13:57, 20 December 2016 by Chgarcia
- The gun is not fully conditioned to 350kV in vacuum, looks like we still have a field emitter to process, despite being up to 365 kV in Krypton:
- A new mechanism for holding the puck on the top heater of the cathode prep chamber has been implemented:
- The previous idea, using a spring-leaf wrapped around the tube of the heater stalk, did not work. The spring did not hold the puck after being heated.
- The new scheme uses sapphire rollers mounted on inconel fingers, the same ones used to grab th puck inside the ball electrode. With this scheme, Mamun and Yan demonstrated the puck is firmly grabbed and pulled towards the heater. The pucks will have to be modified with a groove to make it work. A dummy Al puck was used for the test:
- Cathode Solenoid magnet OSP has been approved: media:operational_safety_procedure_GTS_MLDGT01.pdf. Please read and sign the copy in GTS control room.