Difference between revisions of "October 27, 2020 - Wien magnetic field mapping and characterization meeting"
Latest revision as of 10:36, 29 October 2020
* Use new SCE 20A/75V PS for measurements. This is the same type of PS that will be used for oparations * DC power will install new SCE PS in magnet test ~10/29/20 * Software will provide EPICS controls and hysteresis ramp loops * Label and wire coil leads to ensure positive current produces positive B field per convention * Hysteresis limits: +/- 19.9A
* Hall probe step size 2 mm * On axis values z=[-25,25] cm * Background: measure z=[-30,30] cm. First run the probe in this range w/o the magnet. Then place the magnet for measurements and do a full map at 0 A after hysteresis at 0A . Finally subtract from each value. * Off axis values: limited by 15 mm diameter nickel endplates hole and 6 mm square hall probe. Measure z=[-25,25] cm once at 18A and as far as these limitations allow in +X,-X,+Y,-Y and record X,Y positions for comparing with models. * Measure at +/-19.9A, +/- 15A, +/- 10A, +/-5A and 0A.
CED field map
* Measure B vs. I at center of magnet using the EPICS hysteresis cycle at the beginning then top down through currents.
* V vs I resistance and inductance.