Run Group A
If calling local numbers (with area code prefix 757) from the counting room dial 9+[last 7 digits]. To call other area codes dial 91+[10 digit number]
| valign=top |
Backup system to the On Call:
CLAS12 Run Group A, Spring 2019
Beam energy 10.2 GeV (5 pass)
Important: Document all your work in the logbook!
Remember to fill in the run list at the beginning and end of each run
RC: Nick Markov
- (757) 575-7540
- 9 575 7540 from Counting Room
PDL: Maurizio Ungaro
- (757) 876-1789
- 9 876-1789 from Counting Room
Run Plan: April 8-15, 2019
(Last updated: April 12th, 2019 @ 19:00)
DAQ config: PROD66; Trigger file: rga_v2.trg
Take production data at in 4 hours runs at 50 nA, DAQ config: PROD66; Trigger file: rga_v2.trg.
On the Torus/Magnet power Suppy monitor torus MPS Control Comm light. It should be green. If not, contact Magnet group.
Pay special attention to scalers of the PCAL and FTOF in sector 5 (under scalers/CLAS12 Forward Carriage/Sector 5) and make sure there are no dead channels in PCAL and FTOF. Also, when looking at monitoring histograms, pay special attention to sector 5 for the same detectors. We had a problem with scalers in sector FTOF 4 which went out. Please monitor it very closely (along with sector 5) and if anyhitng happen contact Sergey (DAQ expert) right away.
After the beam is off at 6 am on Monday morning:
<b> At the end of each run, follow the standard DAQ restart sequence
References and Standards:
(last update March 27, 2019 @ 8:53 a.m.)
Restoration of Beam to Hall B
Nominal Beam Positions
Reference Harp Scans for Beam on Faraday Cup: 2H01 
Reference Scalers and Halo-Counter Rates:
SVT acceptable currents: 1 uA
Once Per Day (on day shift):
- Note 1: Be very mindful of the background rates in the halo counters, rates in the detectors, and currents in the SVT for all settings to ensure that they are at safe levels.
The integrated rates on the upstream counters have to be in the range 0-15 Hz (rates up 100 Hz are acceptable) and the rates on the midstream counters have to be in the range 10-20 Hz (acceptable up to 50 Hz) @50 nA. Counting rates in the range of hundreds of Hz may indicate bad beam tune or bleed-through from other Halls.
- Note 2: FTOF HVs: The goal is to minimize the number of power cycles of the dividers.
- should be turned off during the initial beam tuning down to the Faraday Cup after CLAS12 has been off for a long shutdown, when doing a Moller run, when doing harp scans, or if there is tuned/pulsed beam in the upstream beamline.
- should be left on after an initial beam tune has been established and if there are only minor steering adjustments and “tweaks” being made.
- if shift workers have doubts what to do with the HVs, they can always contact the TOF on-call expert for advice.
- Note 3: In case of a Torus and/or Solenoid Fast Dump do the following:
- Notify MCC to request beam OFF and to drop Hall B status to Power Permit
- Call Engineering on-call
- Make separate log entry with copies to HBTORUS and HBSOLENOID logbooks. In the "Notify" field add Ruben Fair, Probir Goshal, Dave Kashy and firstname.lastname@example.org
- Notify Run Coordinator
- Turn off all detectors
- Note 4: When beam is being delivered to the Faraday Cup:
- the Fast Shut Down elements: Upstream, Midstream, Downstream, BOM, and Solenoid should always be in the state UNMASKED
- No changes to the FSD threshold should be made without RC or beamline expert approval
- Note 5: Any request for a special run or change of configuration has to be approved by the RC & documented
- Note 6: Carefully check the BTA every hour and run the script btaGet.py to print for you what HAS TO BE in BTA for this hour. Edit BTA if it is incorrect.
- Note 7: Reset CLAS12MON frequently to avoid histogram saturation. If CLAS12MON is restarted change MAX scale to 10 (default is 5) for DC normalized occupncy histograms. This would prevent them from saturation. See log entry 
- Note 8: Shift workers must check the occupancies! Use this tool to compare to previous runs: 
- Note 9: When you have to turn the whole MVT off, do not press "All HV OFF": Instead go to 'Restore settings' from the MVT overview screen:
- SafeMode.snp for beam tuning and Moeller runs
- MV_HV_FullField.snp for full solenoid field
- MV_HV_MidField.snp wgeb solenoid < 4T
- Note 10: Check the vacuum periodically, make sure vacuum is not higher than 5e-5
- Note 11: Always reset the CFD threshold after all power off/on on the CND CAMAC crate
After the CAMAC crate (camac1) holding the CND CFD boards is switched off for any reason, it is mandatory to reset the associated thresholds typing the following command from any clon machine terminal: $CODA/src/rol/Linux_x86_64/bin/cnd_cfd_thresh -w 0
If this command is failing and the crate is not responding, reboot it as follows: roc_reboot camac1
- Note 12: Shift workers should anyway check routinely scalers to verify they update correctly and make a logbook entry if anomalies are observed after starting a new run.
- Note 13: To solve common SVT issues see https://logbooks.jlab.org/entry/3615766
- Note 14:
For RICH recovery procedures, please see Log entry https://logbooks.jlab.org/entry/3562273. This would apply in the cases of 1. DAQ crash: rich4 is not responding or 2. RICH alarms (LV,missing tile, temperature etc). If it does not work or you are uncertain about what to do, contact the RICH expert on call. Please, note that missing tiles typically occur due to lost communication. Keep in mind that the recovery procedure will kill DAQ. If DAQ is running for other purpose, rather than data taking (for which the RICH acceptance would be important) do not initiate the recovery procedure.
- The most critical parameter for RICH is the temperature of the photosensors. If the temperature rises above limits, an interlock will automatically turn the RICH HV and LV. If this happens, notify the expert on call and keep taking data without RICH.
- Note 15: Asymmetry on FTCal ADC Scalers: If the beam current is 5 - 10 nA, and the observed asymmetry is out of specs, i.e. > +-2.5%, check (1) that the beam position is according to specs, (2) the response of the rest of CLAS (compare the scalers of FTOF, CTOF, ECAL, etc. from the Scalers GUI to the reference for that current, as posted above). If positions and other detectors do not show any anomalies, keep running. The ADC scalers require at least ~15 nA for a reliable response.
- Note 16: If ROC mmft1 is 100% busy: power cycle FTT LV (FT->FTT Overview, HV stays ON), roc_reboot mmft1, wait until it reboots (ssh mmft1 to verify). If the first LV recycle/mmft1 reboot doesn’t work call FT expert